Methodology for identification of liquid concentration in the periodic microstructures applying numerical-experimental laser interferometric methods
Tomas Janušas1 , Giedrius Janušas2 , Arvydas Palevičius3
1, 2, 3Faculty of Mechanical Engineering and Design, Kaunas University of Technology, Kaunas, Lithuania
Vibroengineering PROCEDIA, Vol. 19, 2018, p. 216-220.
Received 20 August 2018; accepted 5 September 2018; published 24 September 2018
33rd International Conference on Vibroengineering in Zittau, Germany, September 24-26th, 2018
The investigation of the high frequency vibrations of the fluid is an important problem in the design of microfluidic devices (periodic microstructures). Laser interferometric methods such as time average holography, high speed double expose holography or laser light diffraction allow to do analysis of high speed fluid flow and dosing or vibration of micro components used in biological and chemical microsystems. Methodology for identification of liquid concentration in the periodic microstructures applying numerical-experimental laser interferometric methods is presented in this paper.
Keywords: microfluidics, microstructures, liquid flow, liquid concentration, holography, diffraction efficiency.
The main component of a lab-on-a-chip is periodic microstructure. Micro channels (periodic microstructures) are used for the fluid transportation, mixing, separation and other processing. The set of micro channels are being formed by soft-lithography , photo polymerisation [2, 3], holographic lithography , laser ablation , micro-injection moulding [6, 7], phase separation , gas foaming  or 3D printing  using a wide range of materials such as polydimethylsiloxane, silicon, polycarbonate, glass, rubber, aluminium and others [11-13]. Usage of new materials such as piezoelectric nanocomposite allows to develop new periodic microstructure with new features like piezoelectric properties or specific optical characteristics. Control methods of micro fluidics by applying acoustic manipulation are known in the world. For this purpose, standing and travelling waves are excited in micro channels . The generated walls of micro hydro dynamical systems are excited by vibration methods which ensure more effective flow of micro fluids . Thin PZT films integrated into micro dynamical systems ensure more effective functionality .
The investigation of the high frequency vibrations of the fluid is an important problem in the design of microfluidic devices. Laser interferometric methods such as time average holography, high speed double expose holography or laser light diffraction allow to do analysis of high speed fluid flow and dosing or vibration of micro components used in biological and chemical microsystems. Therefore, methodology for identification of liquid concentration in the periodic microstructures applying numerical-experimental laser interferometric methods is proposed in this paper.
2. Main equations of holography for phase object investigation
The two-dimensional flow of the ideal compressible liquid in the microchannel (depth is ) is illuminated by laser (wavelength – ). The phase shift of the light of the lasers’ beam which goes through the fluid is :
where is the fluids’ refractive index in static condition and in flow. The refractive index of the liquid in flow condition depends on the dynamic and static densities and could be expressed as:
where correction coefficient should be determined from experiments.
If time average hologram is recorded of that light during a time interval , than phase factor of the reconstructed wave is proportional to:
The intensity of the observed interference pattern is proportional to:
Also, it is possible to use hi speed double exposure holography for the fluid flow analysis if the flow varies steadily and the exposure time is . Than it could be assumed that density field is constant during the exposure. For this case interference pattern is proportional to:
where is the change in phase between the two exposures.
3. Determination of liquids’ concentration
For the development of methodology for identification of liquid concentration in the periodic microstructures applying numerical-experimental laser interferometric methods is selected periodic microstructure (period 4 μm, depth 0.7 μm) made from aluminum and polycarbonate and three liquids (pure water, acetone and glycerol). Optical properties of the selected materials illuminated by red laser ( 633 nm) are presented in Table 1.
Table 1. Optical properties of the materials used in calculations
Polycarbonate (PC) (C16H14O3)
Pure water (H2O)
Optical properties of the microfluidic are investigated by non-destructive optical laser diffractometer (Fig. 1). He-Ne laser diffractometer ( 633 nm) was used in order to register intensities of reflected or transmitted diffraction maxima efficiencies. All liquids are characterized by relative diffraction efficiencies. Relative diffraction efficiency RDE is defined as ratio of intensity of diffracted light to the th diffraction maxima (0, ±1, ±2, …) with intensity of the reflected light from the surface without micro relief . Liquids’ refractive index is determined from comparison of theoretical and experimental diffraction efficiencies. As it is stated in Section 2 liquid density could be calculated from the measured refractive index.
Fig. 1. Measuring schema of diffraction maxima: 1 – sample, 2 – photodiode, 3 – ampere meter, 4 –maxima distribution
The diffraction efficiencies of the microfluidic illuminated by laser are calculated theoretically employing commercial software GSolver V5.2 (Fig. 2). GSolver is based on the modal method. It is set up to work with linear isotropic homogeneous materials and is a full vector implementation of a class of algorithms known as Rigorous Coupled Wave (RCW) analysis.
Fig. 2. Software GSolver V5.2 for calculation of optical parameters of microfluidics
Two periodic microstructure fabricated in aluminum (Al) and polycarbonate (PC) were investigated numerically and experimentally with three liquids (pure water, acetone and glycerol) in order to identify sensitivity of the proposed method. For the measurement of diffraction maxima of the reflected light Al microstructure was used while maxima of the transmitted light were investigated with PC microstructure. Relative diffraction efficiencies of the transmitted and reflected light with three liquids and in air are presented in Fig. 3, 4.
Diffraction efficiencies of the periodic microstructures without an analyte (in air or vacuum) show that periodic microstructure (period 4 μm, depth 0.7 μm) was designed for operation in transmitting mode, i.e. first order diffraction maxima are three times higher in transmitting mode than reflecting. Therefore, periodic microstructure in transmitting mode is more sensitive to refractive index (density) of the analyte. Refractive index change from 1.33 (pure water) to 1.47 (glycerol) leads the decrease of the first order maxima diffraction efficiency four times from 24 % to 6 % (Fig. 4). At the same time reflecting microstructure is not sensitive enough. Results confirm the idea of liquid concentration identification in the periodic microstructures applying numerical-experimental laser interferometric methods.
Fig. 3. Relative diffraction efficiencies of the transmitted light through PC periodic microstructure
Fig. 4. Relative diffraction efficiencies of the reflected light from Al periodic microstructure
Liquids’ refractive index and density in flow condition could be determined from comparison of theoretical and experimental diffraction efficiencies. Change of refractive index by 0.008 increases or decreases diffraction efficiency of the first order diffraction maxima by 1 %. Non-uniform distribution of the fluid density could be determined using holographic methods. Harmonic variation of the fluid density could be registered using time average or high speed double exposure holography.
This research was funded by a grant S-MIP-17-102 from the Research Council of Lithuania.
- Kim P., Kwon K. W., Park M. C., Lee S. H., Kim S. M., Suh K. Y. Soft lithography for microfluidics: a review. Biochip Journal, Vol. 2, Issue 1, 2008, p. 1-11. [Search CrossRef]
- Sun H.-B., Kawata S. Two-photon photopolymerization and 3D lithographicmicrofabrication. Advances in Polymer Science, Vol. 170, 2004, p. 169-273. [Publisher]
- Wu S., Serbin J., Gu M. Two-photon polymerisation for three-dimensional micro-fabrication. Journal of Photochemistry and Photobiology A: Chemistry, Vol. 181, 2006, p. 1-11. [Publisher]
- Stankevičius E., Gedvilas M., Voisiat B., Malinauskas M., Račiukaitis G. Fabrication of periodic micro-structures by holographic lithography. Lithuanian Journal of Physics, Vol. 53, Issue 4, 2013, p. 227-237. [Publisher]
- Hsieh Y. K., Chen S. C., Huang W. L., Hsu K. P., Gorday K. A. V., Wang T., Wang J. Direct micromachining of microfluidic channels on biodegradable materials using laser ablation. Polymers, Vol. 9, Issue 242, 2017, p. 1-16. [Publisher]
- Narijauskaite B., Palevicius A., Narmontas P., Ragulskis M., Janusas G. High-frequency excitation for thermal imprint of microstructures into a polymer. Experimental Techniques, Vol. 37, Issue 5, 2013, p. 41-47. [Publisher]
- Attia U. M., Marson S., Alcock J. R. Micro-injection moulding of polymer microfluidic devices. Microfluidics and Nanofluidics, Vol. 7, Issue 1, 2009, https://doi.org/10.1007/s10404-009-0421-x. [Publisher]
- Liu X., Ma P. X. Phase separation, pore structure, and properties of nanofibrous gelatin scaffolds. Biomaterials, Vol. 30, 2009, p. 4094-4103. [Publisher]
- Salerno A., Oliviero M., Di Maio E., Iannace S., Netti P. Design of porous polymeric scaffolds by gas foaming of heterogeneous blends. Journal of Materials Science: Materials in Medicine, Vol. 20, 2009, p. 2043-2051. [Publisher]
- Silva D. N., Gerhardt De Oliveira M., Meurer E., Meurer M. I., Lopes Da Silva J. V., Santa-Bárbara A. Dimensional error in selective laser sintering and 3D printing of models for craniomaxillary anatomy reconstruction. Journal of Cranio-Maxillo-Facial Surgery, Vol. 36, 2008, p. 443-449. [Publisher]
- Mou L., Jiang X. Materials for microfluidic immunoassays: a review. Advanced Healthcare Materials, Vol. 6, Issue 15, 2017, https://doi.org/10.1002/adhm.201601403. [Publisher]
- Liu X., Lin B. Materials Used in Microfluidic Devices. Encyclopedia of Microfluidics and Nanofluidics. Springer, Boston, MA, 2014. [Publisher]
- Janusas G., Ponelyte S., Brunius A., Guobiene A., Prosycevas I., Vilkauskas A., Palevicius A. Periodical microstructures based on novel piezoelectric material for biomedical applications. Sensors, Vol. 15, Issue 12, 2015, p. 31699-31708. [Publisher]
- Kozuka T., Yasui K. Acoustic manipulation in a microchannel. Proceedings of 20th International Congress on Acoustic, Sydney, Australia, 2010. [Search CrossRef]
- Ogawa J., Kanno I., Kotera H., Wasa K., Suzuki T. Development of liquid pumping devices using vibrating microchannel walls. Sensors and Actuators, Vol. 152, 2009, p. 211-218. [Publisher]
- Cazorla P.-H., Fuchs O., Coschet M., Maubert S., Le Rhun G., Foullet Y., Defay E. Integration of PZT thin films on a microfluidic complex system. IEEE International Ultrasonics Symposium, 2014, p. 491-494. [Publisher]
- Ragulskis M., Palevicius A., Fedaravicius A., Ragulskis L. Applicability of time-average fluid holography for analysis of propagating waves. Optical Engineering, Vol. 44, Issue 10, 2005, p. 1-6. [Publisher]
- Tamulevicius T., Tamulevicius S., Andrulevicius M., Janusas G., Ostasevicius V., Palevicius A. Optical characterization of microstructures of high aspect ratio. Metrology, Inspection, and Process Control for Microlithography XXI, Vol. 6518, 2007. [Publisher]